ELAN is very easy
to use and understand. The following guidelines will help explain its operation.
On: To turn ELAN on, press the power switch on the front of the unit. The beeper will sound once and the led display will indicate a count down sequence from 6 while the sensor is being conditioned for a test.
Ready: When the count down sequence is complete, 00 will illuminate.
Blow: Exhale steadily into the mouthpiece for five seconds. The beeper will sound once to indicate the end of the sample.
Reading: The alcohol level will be indicated on the 3 digit led display. The results will then flash for approximately 5 seconds until the unit turns off automatically.
0ff: To turn ELAN off, press the power switch on the front of the unit The display will indicate OFF and the beeper will sound once.
Low Battery: ELAN will indicate a low battery voltage by displaying the red BATT LOW led.
Sale of the ELAN Personal Alcohol Tester is on the express condition that neither the manufacturer nor the vendor of the device shall be held responsible for the accuracy of any breath teat reading obtained from the device, nor liable for any loss, damage, injury or other consequence resulting, directly or indirectly from the use thereof.
The ELAN Personal
Alcohol Tester must be calibrated periodically to maintain its accuracy. Calibration
frequency requirements depend on usage, and therefore will vary from came to
case. In determining your own requirements, you may wish to start by having
your unit checked for Calibration accuracy on at learnt a quarterly basis. An
immediate calibration check isrecommended if at any time you suspect the unit
may not be giving an accurate reading.
Units in need of service or calibration should be sent to an authorized service center prepaid with an explanation of the problem.
ELAN is warranted against defects in material or workmanship for a period of one year from date of purchase. During the warranty period Alcohol Countermeasure Systems agrees to replace or repair any unit or part that in its judgment is found to be defective, providing the defect is not due to damage or misuse.
|3.50 in x 1.25 in x O.50 in
|±10% @ 100 mg% (.10% MC)
|0C to 40C (30F to 100F)
|3 digit led
5 second continuous display of test results
|Two AAA alkaline battery
|Wait 3 minutes
|Wait 15 minutes after consumption of alcohol
Authorized Service Centers
Countermeasures Systems Corp.
14-975 Midway Boulevard
Mississauga, Ontario Canada L5T 2C6
Countermeasures Systems, Inc
1670 Jasper Avenue, Suite G
Aurora, Colorado, USA 80011
Countermeasure Systems (AA) Pty Ltd
Box 5, 312B North Rocks Road
North Rocks, NSW 2151 Australia
Countermeasure Systems (MB)
Countermeasure Systems is the owner of an extensive number of patents in several
countries covering various aspects of breath sampling procedures and breath
alcohol measurement. This instrument incorporates certain of these breath-testing
principles as outlined and protected by these patents.
ELITE is a registered trademark of Alcohol Countermeasure Systems
© 2000 Alcohol Countermeasure Systems
Home | Breath
Alcohol Testing | Drug
Testing | Accessories | Paternity Testing Services | FAQ's | Links
Guest Register | View Cart | Contact Us
Breath Alcohol Testing Spec Sheets:
Ensure Model J7 | Snap® Disposable | DRIVESAFE Personal Alcohol Tester
Elan Portable BAT | QED Saliva Disposable Alcohol Test | Alert Model J4X | Alert J4 ECX Evidential Portable
Scotty Dry Gas Kit | Guth Wet Bath Kit | ALCOSCAN | Ethyl Alcohol Solutions
Drug Testing Spec Sheets:
Hair Testing | InstaStick Drug Test Strips | Instacheck Drug Screen | InstaCup Drug Screen | Intercept Oral Fluid Drug Test